Find great deals for HP a Spectrum Analyzer Display Mainframe W/ HP b Pattern Generator. Shop with confidence on eBay!. Find great deals for Hewlett Packard HP a Display With b Optical Spectrum Analyzer. Shop with confidence on eBay!. Software and documentation supplied. This service guide is part of an Option OB3 package which includes: HP A Service Guide. HP A Component.
||7 August 2016
|PDF File Size:
|ePub File Size:
||Free* [*Free Regsitration Required]
Knowing I had nothing to loose, I went and 70004s the transformer can with a Dremel, and removed the resin with hot air and tweezers, carefully to not damage the windings. I used a thin teflon tube and a copper wire to recreate the four turns that form the secondary and guess what, it worked!!
You’ll probably want to use an isolation transformer and differential probes, and keep one hand on your pocket at all times.
I built this “adapter” thing as a temporary solution, you too know what that means: There is a current sense resistor and this is what current looked at turn-on try:. You can see the power supply slowly increasing the current at turn on until it gets too high before it starts regulating the voltage, and quits.
Showing Results for Keysight Technologies (Agilent HP) 70004A
np Many suggested changing every capacitor in the board, something I was not going to do, I wanted to find h the fault really was all capacitors turned to be ok:. This last one required some improvisation because it is a quad-transistor instead of dual, and the DIP package is very different. Could this be fixed? I confessed before that I killed a few components on the process, and I could not find replacements for some of them: I did not know where to start.
The supply operates a high switching frequency and many parts of the circuit are shorted at DC, so I could not find this until I took both transformers out. I could see the power LED flash briefly upon turn-on and then Something was drawing more current than expected and the current-limit protection was tripping.
A Color System Display [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
The block diagram was invaluable, this switching power supplies are complex. And a final picture of the twin transformers in the working power supply. One day, it refused to turn on. This is the beast on the bench, that PCB looks very up to me.
Can you tell one from the other? I was happy to find it actually worked You really need to know what you are doing before probing and testing a thing like this live. No surpise, the secondary jacket was burned and touching the primary in one spot:.
Keysight Technologies (Agilent HP) A used or new for sale at used-line
This board is a latter revision than the ones that appear on the schematics but there are not many changes. This paid for itself in just this repair. Do you see that suspicious potting that squeezed outside the can in one of them?:. After many hours of testing and probing, and and few components that exploded in the process most times my fault I found there was a transformer with a 14 ohm leak between primary and secondary.
Home Projects Music Photo.
Hewlett Packard HP 70004a Display With 70950b Optical Spectrum Analyzer
I bought this cheap A color display as faulty. There is no way to check one of the blocks of the power supply alone, you have to give it power and hope for the best. High voltage, big and the troubleshooting will need to be done live.
I was lucky to have bought a FLUKE D differential voltage probe this year at Friedrichshafen to the nice guy with grey hair and glasses that sells mostly Fluke stuff.